The carbon-silicon analyzer in front of the furnace determines that the content of molten iron is a unique place of the thermal analysis instrument, and any other component analysis method cannot be substituted. The fast carbon and silicon analyzer in front of the furnace can detect the molten iron analyzer, which is also an advantage that cannot be compared with spectral analysis and chemical analysis.
Furnace carbon and silicon analyzer installation
The meter should be installed in a place with relatively low temperature and relatively little dust. Do not place the meter in direct sunlight. Otherwise, it will affect the service life and test accuracy of the meter.
The carrier on which the meter is installed should be stable to prevent the meter from falling.
After connecting the thermal analysis support and connecting the power, please warm up the instrument for more than 10 minutes to test.
When the fixed carbon cup is placed on the analysis support, the green light is on to indicate that the system is ready for measurement. After the molten iron is poured into the cup, the yellow light is on (the green light is off), the meter starts to measure the temperature cooling curve, and different reference signals are displayed according to the parameters set by the meter. The thermal analyzer calculates CEL and C% according to the preset procedure , Si% equivalent. After detection and calculation, the red light is on (yellow light is off). After the red light goes out, you can press the corresponding button to read the corresponding measured value.
The status of the entire measurement process is indicated by green lights, yellow lights, and red lights, which are turned on until they are all off.
Front-of-furnace carbon-silicon analyzer safety guide
Warning: To prevent accidents, please do not expose the analyzer to rain or humidity, or place it on an unstable surface; please do not open the case by yourself. If you need repair, please ask our company's after-sales service personnel. Or, under the guidance of our company's after-sales service personnel, open the case for maintenance and parameter adjustment.
This instrument uses AC 200V / 50Hz power supply. Uses a standard three-pin power plug. For safety, please ground the socket ground terminal reliably.
Improper grounding will cause electric shock or inaccurate measurements!
Do not use a damaged or loose power outlet and wires!
Please do not pull the power cord to unplug the power plug, and do not pull or plug the power plug with wet hands!
If the meter is not used for a long time, disconnect the power plug!